Mode: HDCG I-2400kV/20mA
IEC 61643-1 2011 Surge Protective Devices Connected to Low-voltage Power Distribution Systems-Part 1: Performance and Testing Methods
IEC 61083-1 Instruments and Software Used for Measurement in High-voltage Impulse Tests-Part 1: Requirements for Instruments
IEC 61083-2 Instruments and Software Used for Measurement in High-voltage and High-current Tests-Part 2: Requirements for Software for Tests with Impulse Voltages and Currents
IEC 62475 High Current Test Techniques-Definitions and Requirements for Test Current and Measuring Systems
Rated Voltage: 2400kV
Rated Current : 20mA
Ripple Factor: <3%(20mA resistive current )
Voltage stability : Test object relative voltage drop <1% under resistive currentUC5,C6,C7,C8=0.16μF(DC) U=600kV ; UC2,C3,C4=0.16μF (AC) U=600kV
Forward voltage drop: Less than 8000VHigh-voltage silicon rectifier :Forward voltage drop: Less than 8000V
Nominal peak reverse voltage :UK=600kV (8stages)
Rated current:2000mA IFSM (10ms):20A
Forward current rating :20mA
Forward voltage drop: Less than 4000V
Working time: one hour on, one hour off, 4 cycle per day.
Main circuit structure, this system is designed as unilateral unsymmetrical single stage voltage-multiplying circuit, besides test transformer, voltage regulator, the filter capacitor, multivoltage capacitor, silicon rectify stack is placed separately on three basement. The resistance voltage divider and earthing switch has the same basement which equip universal wheel and lifting jack. We can customize movable and stable type. Manual change polarity. The top electrode(corona ring) is special designed for DC generator. See outside drawing. There are safety earthing system on the generator, which protects the safety of staff and equipment.
HIMALAYAL provides innovative products that comprise high voltage test equipment, full-set high voltage test instruments, on-site test systems and on-line monitoring systems, and services regarding tests, diagnostics and asset monitoring for clients in power industry. We help to ensure safety and reliability in the power generation, transmission and distribution.